How to setup the LiDCO hemodynamic monitor for non invasive monitoring in a few simple steps.
Step 1: Power on the LiDCO monitor
• Connect the power supply (optional)
• Power on using the button underneath the monitor
* when A/C power is not available, the monitors battery can offer 3hrs of operation when fully charged.
Step 2: Select Rapid & CNAP
• Select “R” for the LiDCOrapid trend of Hemodynamics, and
• Select “C” for CNAP (Continuous Non-Invasive Arterial Pressure) from the finger
• The wedge will turn from blue to green when selected.
• Press the white check mark in the center of the wheel to confirm and move
Step 3: Enter patient details
• Enter Pt. ID number, age, height, weight, gender and patient type
• Press the green forward arrow* to enter monitoring mode
*If the arrow isn’t green, go back to complete all details
Step 4: Attach CNAP to patient
• Place the patient’s index finger onto the sizing gauge on top of the cuff controller to determine the correct cuff size (S/M/L).
*the cuffs should slide easily over the knuckles without resistance
• Position the finger cuffs at the base of the pointer, middle and ring finger
• Connect finger cuffs to cuff controller (black) & cuff controller to monitor cable (orange)
• Fasten cuff controller to forearm with strap provided
*if the arm will be tucked during surgery, ensure the patient is padded to avoid any potential compression injuries.
Step 5: Begin signal acquisition sequence
• If possible, ensure the Cuff controller is at the level of the Right Atrium
• Press the green button in the Blood Pressure Channel to start the CNAP signal acquisition
*if the arrow remains grey, check all connections, and observe the area at the top of the screen for troubleshooting messages.
Step 6: Calibrate with most recent Arm Cuff measurement
Enter values for systolic and diastolic for NIBP when prompted
Step 7: View perfusion index
• This indicates the relative strength of perfusion in the finger. You need at least one green box for the measurement to proceed.
• The further to the right the green box is, the stronger the pulse signal.
Step 8: View CNAP finger signal acquisition
• One finger cuff will inflate and deflate several times as the monitor senses the patients pulse. Observe as the red line on the pressure graph moves slowly up and down as the finger cuff inflates and deflates.
• Next, the finger cuff will begin to counter-pulse and display an arterial waveform (~90 seconds total)
• The CNAP has successfully detected the blood pulsatility in the finger.
Step 9: View ALL trended values
• Observe NIBP arm cuff measurement used for calibrating the LiDCO finger sensors appears in parenthesis below the current finger cuff pressure measurement
• Observe continuous trending of all measured parameters: CO, CI, SV, SVR, SVV, PPV, & Event response % Change.
Click on the links to find out how to give a fluid challenge and how to setup minimally invasive hemodynamic monitoring
Got a question? Need some help?
Call us now on +44 (0) 207 749 1500 or email us at email@example.com and we’ll get back to you ‘stat!’